To obtain a carry case, order Agilent part number A. Page 0 and To view the sensor calibration tables currently stored in the power meter, press. This will ensure that you obtain the most complete and accurate service information. This softkey also accesses the Battery menu which allows you to set the display backlight to On, Off or Timed mode.
|Date Added:||27 August 2009|
|File Size:||40.51 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Ratio measurements can be displayed relative to a stored reference.
Press this softkey to display the measurement results in dBm. Observe static precautions while removing and replacing the battery module.
E4418B and E4419B RF Power Meter User’s Guide
In this instance, channel A averaging is calculated with a resolution of four and channel B with a resolution of two. Page 73 3 To confirm your choice, press the appropriate softkey.
Exclusive Remedies Do not proceed beyond a caution until the indicated conditions are fully understood and met. Power Zgilent Specifications This parameter is also known as short term stability and is specified as the change in the power meter indication over a short time interval usually one minute for a constant input power at a constant temperature.
Comments to this Manuals Your Name.
The display now shows only the digital window. General Characteristics Binding post, accepts 4 mm plug or bare wire connection. TTL logic level output when a measurement exceeds a predetermined limit. Channel Measurements Single channel measurements can be displayed relative to a stored reference.
Agilent EB GHz EPM Series Dual Channel Power Meter 90 Day | eBay
Do not dispose of by burning. The keys move the position of the cursor. The following procedure details agiletn to make a measurement on channel A. Press this softkey to switch step detection on or off.
Wait Symbol This pop-up window is displayed when you are required to modify numeric, or alphanumeric data. This key allows you to choose either a one or a two window display. Error Messages — Exponent too large A numeric parameter was found whose exponent was larger than 32, Entering a calibration factor using this softkey is only possible when you are using an Agilent Series power sensor or N Series power sensor with Option CFT.
Agilent E4419B 110 GHz EPM Series Dual Channel Power Meter 90 Day
The agillent menu structured to this key affects the currently selected window. The stops the display and returns you to the previous menu. Step detection reduces the settling time of the filter by re-initialising the filter after a significant step increase or decrease has been detected in the measured power. The current setting of the resolution is highlighted on the softkey. Page 83 If you measure a signal with a frequency outside the frequency range defined in the frequency dependent offset e4419bb, the power meter uses the highest or lowest frequency point in the frequency dependent offset table to calculate the offset.
The measurement windows can display the result in agjlent a digital or analog format or both as shown in the following figures.
HP Agilent Keysight, EB
Page 58 30 dB reference attenuator and an APC 2. Rel Off On Successive measurements are now displayed relative to the reference value.
Got it, continue to print. Editing Frequency Dependent Offset Tables Use theand keys to move between the table name and its frequencies and offsets. Confirmation Window This pop-up window is displayed when you are required to press Confirm verify your e44419b selection.
To confirm your choice, press the appropriate measurement units.
Contacting Agilent Technologies If you have a problem with your power meter, first refer to the section. Page if enabled at the rear ayilent TTL outputs. This allows you to measure the power of a pulsed signal.