Yes you can do this, but in principle you are writing your own driver. Klomp 1, 1 7 No offense was taken. You can use an instrument driver for a particular instrument as is. Email Required, but never shown.
|Date Added:||5 February 2012|
|File Size:||24.80 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Agilent 34401 Meter
Most instrument drivers have menu palettes that include the following components. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Agilent A time log RS Has a 3 Application Examples and a “Getting Started” one which should get you on your way. Feel 34401s to write us at les. Get acquainted with sub-vi’s. In the following figure we have opened the Agilent Read Multiple Measurements.
I found the function you suggested and it’s indeed a more elegant solution. Regarding your question just coding your own program using GPIB.
Keysight Instrument Drivers | Keysight (formerly Agilent’s Electronic Measurement)
Usually the drivers are accessible and you will see all the GPIB calls that are made by the driver. Sign up using Email lavbiew Password.
Message 6 of 7. Sign up using Facebook. Unfortunately I haven’t been able to find a way to achieve this.
Each VI corresponds to a programmatic operation, such as configuring, reading from, writing to, and triggering an instrument. Email Required, but never shown. Numerous Configuration VIs can exist, depending on the particular instrument.
Message 2 of 7. Message 6 of 6. Most Active Hardware Boards: Don’t be afraid of experimenting.
The Data VIs- transfer data to or from the instrument. Open a blank vi. I’m not sure about instructions, but if you installed the driver and opened the library, you will get sub-vi’s which have a particular application purpose.
No offense was lsbview. Yes you can do this, but in principle you are writing your own driver. Sign up or log in Sign up using Hpp. I get an error windows with the values the multimeter has acquired, I think. Most Active Software Boards: You can create instrument control 3440a and systems by programmatically linking instrument driver VIs on the block diagram. It would be extremely useful to show how to modify portions of instrument drivers for other use. Then the Configure Measurement VI is called to set the instrument up for the desired measurement.
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Both of these drives are supported by Agilent and come complete with example code for common tasks. The Configuration VIs- a collection of software routines that configure the instrument to perform the operation you want.